Background: As advances in neuroimaging further our understanding of the brain's functional connectivity, neuropsychology has moved away from a regional approach of attributing behavior to a specific...Read More
This is BrainsWay’s global website. The global website is not intended for persons in the United States and includes information on clinical indications that were not cleared by the FDA, which are subject to further US regulatory review for safety and efficacy. BrainsWay D is cleared by the FDA only for patients with MDD who failed to respond to one or more anti-depressants in the current episode, and for patients with OCD as an adjunct treatment.
Journal: Journal of Clinical Neurophysiology 24(1):31-38 (2007)
Authors: Y Roth, A Amir, Y Levkovitz, A Zangen
The H-coils are a novel development in transcranial magnetic stimulation (TMS), designed to achieve effective stimulation of deep neuronal regions without inducing unbearable fields cortically, thus broadly expanding the potential feasibility of TMS for research and for treating various neurologic disorders.
To compare the field distribution of two H-coil versions, termed H1 and H2, and of a standard figure-8 coil.
Three-dimensional electrical field distributions of the H1 and H2-coils, designed for effective stimulation of prefrontal regions, and of a standard figure-8 coil, were measured in a head model filled with physiologic saline solution.
With stimulator output at 120% of the hand motor threshold, supra threshold field is induced by the H1-coil at lateral and medial frontal regions at depths of up to 4 to 5 cm, and by the H2-coil at medial prefrontal regions up to 2 to 3 cm, and at lateral frontal regions up to 5 to 6 cm. The figure-8 coil induced supra threshold field focally under the coil’s central segment, at depths of up to 1.5 cm.
The ability of the H-coils to stimulate effectively deeper neuronal structures is obtained at the cost of a wider electrical field distribution in the brain. However, the H-coils enable simultaneous stimulation of several brain regions, whereas the depth penetration in each region can be controlled either by adjusting the stimulator output, and/or by varying the distance between various coil elements and the skull.